*Also, see sections on Cardiac Monitors, such as the Reveal and Confirm Products
Cardiac Pacemakers and Implantable Cardioverter Defibrillators
Cardiac pacemakers and implantable cardioverter defibrillators (ICDs) are crucial implanted devices for patients with heart conditions and serve to maintain quality of life and substantially reduce morbidity. Expanded indications for cardiac pacemakers and ICDs (e.g., heart failure, obstructive sleep apnea, and other conditions) emphasize that an increasing number of patients will be treated with these devices.
Currently, these cardiovascular implants are considered a relative contraindication for patients referred for MR procedures. Additionally, individuals with cardiac pacemakers and ICDs should be prevented from entering the MR environment because of potential risks.
MR conditional cardiac pacing systems are now available for patients referred for MR examinations and a comprehensive review of this topic has recently been presented in the Journal of Cardiovascular Magnetic Resonance (JCMR) by Shinbane, Colletti, and Shellock (Magnetic resonance imaging in patients with cardiac pacemakers: Era of MR conditional design. Journal of Cardiovascular Magnetic Resonance 2011;197:W457-9).
For additional information on the topic of cardiac pacing systems and MRI, please refer to the following sections, including the sections that refer to “MR Conditional” cardiac pacing systems: Cardiac Pacemaker: Accent MRI Pacemaker System, Cardiac Pacemaker: Advisa DR MRI SureScan Pacing System, Cardiac Pacemaker: Ensura DR MRI SureScan Pacing System, Cardiac Pacemaker: EnRhythm MRI SureScan Pacing System, Cardiac Pacemaker: Revo MRI SureScan Pacing System, Cardiac Pacemakers: ProMRI Pacing Systems, and Extremity MR System.
Cardiac pacemakers and ICDs have been suggested to present potential problems to patients undergoing MR procedures from various mechanisms, including:
1) Movement and/or vibration of the pulse generator or lead(s)
2) Temporary or permanent modification of the function (i.e., damage) of the device
3) Inappropriate sensing, triggering, or activation of the device
4) Excessive heating of the leads
5) Induced currents in the leads
6) Electromagnetic interference
The effects of the MR environment and MR procedures on the functional and operational aspects of cardiac pacemakers and ICDs vary, depending on several factors including the type of device, how the device is programmed, the static magnetic field strength of the MR system, and the imaging conditions used for the procedure (i.e., the anatomic region imaged, type of transmit RF coil used, the pulse sequence, amount of radiofrequency energy used, etc.). Notably, most of the data concerning the deleterious effects of MR imaging on cardiac pacemakers involved the use of older versions (i.e., pre-1996) of these devices.
Recently, many compelling reports have been published concerning “modern-day” pacemakers (i.e. devices with decreased ferromagnetic components and more sophisticated circuitry) that indicate that certain patients may undergo MR examinations without harmful effects by following specific guidelines to minimize or prevent risks. Thus, there is growing evidence that it may be possible to perform MR procedures safely in patients with cardiovascular devices under highly controlled conditions (see below). Alternatively, as previously mentioned, MR conditional cardiac pacing systems now exist that were specifically designed to provide a high margin of safety for patients referred for MR procedures.
Implantable cardioverter defibrillators (ICDs) are medical devices designed to automatically detect and treat episodes of ventricular fibrillation, ventricular tachycardias, bradycardia, and other conditions. When a problem is identified, the device can deliver defibrillation, cardioversion, antitachycardia pacing, bradycardia pacing, or other therapy.
In general, exposure to an MR system or to an MR procedure has similar effects on an ICD as those previously described for a cardiac pacemaker, since some of the basic components are comparable. However, there are several unique aspects of ICDs that impact the possible safe performance of MR procedures in patients with these devices. Therefore, patients and individuals with these devices are generally not allowed to enter the MR environment. In addition, since ICDs also have electrodes placed in the myocardium, patients are typically not permitted to undergo MR examinations because of the inherent risks related to the presence of these conductive materials.
Similar to cardiac pacemakers, it is anticipated that, in lieu of developing a truly acceptable ICD for the MR environment, safety criteria may be determined for “modern-day” ICDs that entail using specialized programming, monitoring procedures, and MR conditions to allow patients to undergo MR examinations safely. Several recent studies have described patients with ICDs examined by MR imaging without serious problems with the caveat that ICDs, in fact, tend to be more problematic than pacemakers relative to the use of MR examinations. Notably, potential problems remain even for “modern” cardiac pacemakers and ICDs as indicated by reports by Gimbel (2009) and Mollerus et al. (2009), such that extreme caution must be exercised when scanning patients with these devices.
Growing Evidence for Performing MR Procedures in Nonpacemaker Dependent Patients
Harmful effects related to performing MRI procedures in patients with cardiac pacemakers have been documented. Notably, in virtually every case involving a fatality, the patient apparently entered the MR environment without the staff knowing a cardiac pacemaker was present. Importantly, these deaths were poorly characterized, no electrocardiographic data were available for review, it was unknown whether these patients were pacemaker dependent, and the actual cause or mechanism of death were not confirmed. By comparison, no irreversible harm has been reported when patients with cardiac pacemakers were carefully monitored during MR procedures and/or the devices underwent reprogramming prior to the scans.
Regardless of the known hazards of subjecting a patient with a cardiac pacemaker to the MR environment, numerous patients (1,000+) have now undergone MR imaging during purposeful, monitored procedures performed in order to conduct necessary diagnostic examinations. These patients were safely and successfully imaged using MR systems operating at static magnetic fields ranging from 0.2-Tesla to 3-Tesla without serious adverse events.
In consideration of the above, there is increasing evidence from in vitro, laboratory, and clinical studies that strict restrictions prohibiting MR procedures in patients with “modern” cardiac pacemakers and ICDs may be modified. Similar to performing MR procedures in patients with other electronically-activated devices (e.g., bone fusion stimulators, cochlear implants, neurostimulation systems, programmable injusion pumps, etc.), scanning patients with cardiac pacemakers and ICDs involves following highly specific procedures to ensure patient safety.
Interestingly, a recent study by Gimbel (2008) reported scanning patients with cardiac pacemakers at 3-Tesla with no restrictions placed on pacemaker dependency, region scanned, device type, or manufacturer. This somewhat limited experience suggested that patients may undergo carefully tailored 3-Tesla MRI scans when pre-MRI reprogramming of the device occurs in conjunction with extensive monitoring, supervision, and follow-up.
Proposed Guidelines for Performing MRI in Nonpacemaker Dependent Patients
Guidelines have been presented by Martin et al. (2004), Roquin et al. (2004), Loewy et al. (2004), the American College of Radiology (2007) and the American College of Cardiology/American Heart Association (2007) with regard to performing MR procedures in nonpacemaker dependent patients. These guidelines include, the following:
-Establish a risk-benefit ratio for the patient
-Obtain written and verbal informed consent
-Pretest pacemaker functions using appropriate equipment outside of the MR environment
-A cardiologist/electrophysiologist should decide whether it is necessary to program the pacemaker prior to the MR examination
-A cardiologist/electrophysiologist with Advanced Cardiac Life Support (ACLS) training must be in attendance for the entire MRI examination
-The patient should be monitored continuously during the MR procedure (e.g., blood pressure, pulse rate, oxygen saturation, and ECG)
-Appropriate personnel, a crash cart, and de?brillator must be available throughout the pro¬cedure to address an adverse event
-Maintain visual and voice contact throughout the procedure with the patient
-Instruct the patient to alert the MR system operator of any unusual sensations or problems so that, if necessary, the MR system operator can immediately terminate the procedure
-After the MRI examination, a cardiologist/electrophysiologist should interrogate the pacemaker to con?rm that the function is consistent with the pre-examination state
MRI at 1.5 and 3-Tesla and Cardiac Pacemakers and ICDs: Magnetic Field Interactions
As previously discussed, one important safety aspect of the MR environment on cardiac pacemakers and ICDs is related to magnetic field interactions. Component parts of pacemakers and ICDs, such as batteries, reed-switches, or transformer core materials may contain ferromagnetic materials. Therefore, substantial magnetic field interactions may exist, causing these implants to move or be uncomfortable for patients or individuals. Therefore, as an important part of evaluating pacemakers and ICDs, tests for magnetic field interactions have been conducted using MR systems operating at static magnetic field strengths ranging from 0.2-Tesla (i.e., the dedicated-extremity MR system) to 3-Tesla.
Luechinger et al. investigated magnetic field interactions for thirty-one cardiac pacemakers and thirteen ICDs in association with exposure to a 1.5-Tesla MR system (Gyroscan ACS NT, Philips Medical Systems, Best, The Netherlands). The investigators reported that “newer” cardiac pacemakers had relatively low magnetic force values compared to older devices. With regard to ICDs, with the exception of one newer model (GEM II, 7273 ICD, Medtronic, Inc., Minneapolis, MN), all ICDs showed relatively high magnetic field interactions. Luechinger et al. concluded that modern-day pacemakers present no safety risk with respect to magnetic field interactions at 1.5-Tesla, while ICDs could pose problems due to strong magnet-related mechanical forces.
The clinical use of 3-Tesla MR systems for brain, musculoskeletal, body and cardiovascular applications is increasing. Because previous investigations performed to determine MR safety for pacemakers and ICDs used MR systems with static magnetic fields of 1.5-Tesla or less, it is crucial to perform ex vivo testing at 3-Tesla to characterize magnetic field-related safety for these implants, with full appreciation that additional MR safety issues exist for these devices, as described-above.
An important aspect of determining magnetic field interactions for metallic implants involves the measurement of translational attraction. Translational attraction is assessed for metallic implants using the standardized deflection angle test recommended by the American Society for Testing and Materials (ASTM) International. According to ASTM International guidelines, the deflection angle for an implant or device is generally measured at the point of the “highest spatial gradient” for the specific MR system used for testing. Notably, the deflection angle test is commonly performed as an integral part of safety testing for metallic implants and devices.
Various types of magnets exist for commercially available 1.5- and 3-Tesla MR systems, including magnet configurations that are used for conventional “long-bore” scanners and newer “short-bore” systems. Because of physical differences in the position and magnitude of the highest spatial gradient for different magnets, measurements of deflection angles for implants using long-bore vs. short-bore MR systems can produce substantially different results for magnetic field-related translational attraction, as reported by Shellock et al.
The implications are primarily for magnetic field-related translational attraction with regard to long-bore vs. short-bore 3-Tesla MR systems (with short-bore scanners producing greater translational attraction for a given implant). Therefore, a study was conducted on fourteen different cardiac pacemakers and four ICDs to evaluate translational attraction for these devices in association with long-bore and short-bore 1.5- and 3-Tesla MR systems. Deflection angles were measured based on guidelines from the ASTM International.
In general, deflection angles for the cardiovascular implants that underwent evaluation were significantly (p<0.01) higher on 1.5- and 3-Tesla short-bore scanners compared to long-bore MR systems. For the 1.5-Tesla MR systems, three cardiac pacemakers (Cosmos, Model 283-01 Pacemaker, Intermedics, Inc., Freeport, TX; Nova Model 281-01 Pacemaker, Intermedics, Inc., Freeport, TX; Res-Q ACE, Model 101-01 Pacemaker; Intermedics, Inc., Freeport, TX) exhibited deflection angles greater than 45 degrees (i.e., exceeding the recommended ASTM International criteria) on both long-bore and short-bore 1.5-Tesla MR systems. The findings indicated that these devices are potentially problematic for patients from a magnetic field interaction consideration.
With regard to the 3-Tesla MR systems, seven implants exhibited deflection angles greater than 45 degrees on the long-bore 3-Tesla scanner, while 13 exhibited deflection angles greater than 45 degrees on the short-bore 3-T MR system (refer to The List for information on the cardiac pacemakers and ICDs that underwent testing). Importantly, the findings for magnetic field-related translational attraction were substantially different comparing the long-bore (i.e., lower values) and short-bore (i.e., higher values) MR systems.
As stated, other factors exist that may impact MRI issues for these cardiac pacemakers and ICDs. Therefore, regardless of the fact that magnetic field interactions may not present a risk for some of the cardiovascular implants that have been tested, these potentially hazardous mechanisms must be considered carefully for these devices.
MR Conditional Cardiac Devices
As a result of considerable design changes, several MR Conditional cardiac pacing systems and implantable cardioverter defibrillators (ICDs, as well as cardiac monitors, are now available for patients referred for MR examinations including those manufacturered by Medtronic, Inc., Sorin, Boston Scientific, Biotronik, and St. Jude Medical. For additional information pertaining to these cardiac devices and MRI, please refer to the pertinent sections in this textbook and visit the websites of the aforementioned manufacturers.
Important Note to MR Healthcare Professionals: Different MRI guidelines may exist for different countries. Therefore, use the appropriate information for your country.
MR healthcare professionals are advised to contact the respective manufacturer in order to obtain the latest safety information to ensure patient safety relative to the use of an MR procedure.
Acha MR, et al. Increased perforation risk with an MRI-conditional pacing lead: A single-center study. Pacing Clin Electrophysiol 2015;38:334-42.
Achenbach S, et al. Effects of magnetic resonance imaging on cardiac pacemakers and electrodes. Am Heart J 1997;134:467-473.
Ahmed FZ, et al. Not all pacemakers are created equal: MRI conditional pacemaker and lead technology. J Cardiovasc Electrophysiol 2013;24:1059-65.
Ainslie M, et al. Cardiac MRI of patients with implanted electrical cardiac devices. Heart 2014;100:363-9.
Alagona P, et al. Nuclear magnetic resonance imaging in a patient with a DDD pacemaker. Pacing Clin Electrophysiol 1989;12:619 (letter).
Al-Wakeel N, et al. Cardiac MRI in patients with complex CHD following primary or secondary implantation of MRI-conditional pacemaker system. Cardiol Young 2015 Epub ahead of print.
American Society for Testing and Materials (ASTM) Designation: F2052. Standard test method for measurement of magnetically induced displacement force on passive implants in the magnetic resonance environment. West Conshohocken, PA.
Anfinsen OG, et al. Implantable cardioverter defibrillator dysfunction during and after magnetic resonance imaging. Pacing Clin Electrophysiol 2002;25:1400-1402.
Awad K, et al. Clinical safety of the Iforia ICD System in patients subjected to thoracic spine and cardiac 1.5T MRI scanning conditions. Heart Rhythm 2015 Epub ahead of print
Baikoussis NG, et al. Safety of magnetic resonance imaging in patients with implanted cardiac prostheses and metallic cardiovascular electronic devices. Ann Thorac Surg 2011;91:2006-11.
Bailey SM, et al. Magnetic resonance imaging (MRI) of the brain in pacemaker dependent patients. Pacing and Clinical Electrophysiology 2005;2:S128.
Bailey WM, et al. Clinical safety of the ProMRI pacemaker system in patients subjected to thoracic spine and cardiac 1.5-T magnetic resonance imaging scanning conditions. Heart Rhythm 2015 Epub ahead of print.
Barbier T, et al. An RF-induced voltage sensor for investigating pacemaker safety in MRI. MAGMA 2014;27:539-49.
Baser K, et al. High ventricular lead impedance of a DDD pacemaker after cranial magnetic resonance imaging. Pacing Clin Electrophysiol 2012;35:e251-3.
Bhachu DS, Kanal E. Implantable pulse generators (pacemakers) and electrodes: Safety in the magnetic resonance imaging scanner environment. J Magn Reson Imag 2000;12:201-204.
Bhandiwad AR, et al. Cardiovascular magnetic resonance with an MR compatible pacemaker. J Cardiovasc Magn Reson 2013;15:18.
Boilson BA, et al. Safety of magnetic resonance imaging in patients with permanent pacemakers: A collaborative clinical approach. J Interv Card Electrophysiol 2012;33:59-67
Bonnet CA, Elson JJ, Fogoros RN. Accidental deactivation of the automatic implantable cardioverter defibrillator. Am Heart J 1990;3:696-697.
Bovenschulte H, et al. MRI in patients with pacemakers: Overview and procedural management. Dtsch Arztebl Int 2012;109:270-5.
Buendía F, et al. Nuclear magnetic resonance imaging in patients with cardiac pacing devices. Rev Esp Cardiol 2010;63:735-9.
Buendía F, et al. Cardiac magnetic resonance imaging at 1.5 T in patients with cardiac rhythm devices. Europace 2011;13:533-8.
Burke PT, et al. A protocol for patients with cardiovascular implantable devices undergoing magnetic resonance imaging (MRI): Should defibrillation threshold testing be performed post-(MRI). J Interv Card Electrophysiol 2010;28:59-66.
Calcagnini G, et al. In vitro investigation of pacemaker lead heating induced by magnetic resonance imaging: Role of implant geometry. J Magn Reson Imag 2008;28:879-86.
Chow GV, et al. MRI for patients with cardiac implantable electrical devices. Cardiol Clin 2014;32:299-304.
Cohen JD, et al. Determining the risks of magnetic resonance imaging at 1.5 Tesla for patients with pacemakers and implantable cardioverter defibrillators. Am J Cardiol 2012;110:1631-6.
Colletti PM, Shinbane JS, Shellock FG. "MR-conditional" pacemakers: The radiologist's role in multidisciplinary management. Am J Roentgenol 2011;197:W457-9.
Corzani A, et al. Clinical management of electromagnetic interferences in patients with pacemakers and implantable cardioverter-defibrillators: Review of the literature and focus on magnetic resonance conditional devices. J Cardiovasc Med 2015 ;16:704-13.
Del Ojo JL, et al. Is magnetic resonance imaging safe in cardiac pacemaker recipients? Pacing Clin Electrophysiol 2005;28:274-8.
Duru F, et al. Pacing in magnetic resonance imaging environment: Clinical and technical considerations on compatibility. Eur Heart J 2001;22:113-124.
Duru F, Luechinger R, Candinas R. MR imaging in patients with cardiac pacemakers. Radiology 2001;219:856-858.
Erlebacher JA, Cahill PT, Pannizzo F, Knowles RJR. Effect of magnetic resonance imaging on DDD pacemakers. Am J Cardio 1986;57:437-440.
Expert Panel on MR Safety, Kanal E, Barkovich AJ, Bell C, et al. ACR guidance document on MR safe practices: 2013. J Magn Reson Imag 2013;37:501-30.
Fanourgiakis J, Kanoupakis E. Cardiac rhythm management devices in a magnetic resonance environment. Expert Rev Med Devices 2014;11:199-203.
Faris OP, Shein M. Food and Drug Administration perspective: Magnetic resonance imaging of pacemaker and implantable cardioverter-defibrillator patients. Circulation 2006;114:1232-1233.
Ferreira AM, et al. MRI-conditional pacemakers: Current perspectives. Med Devices 2014;7:115-124.
Fetter J, et al. The effects of nuclear magnetic resonance imagers on external and implantable pulse generators. Pacing Clin Electrophysiol 1984;7:720-727.
Fiek M, et al. Complete loss of ICD programmability after magnetic resonance imaging. Pacing Clin Electrophysiol 2004;27:1002-4.
Forleo GB, et al. Clinical and electrical performance of currently available MRI-safe pacing systems. Do all devices perform in the same way? Int J Cardiol 2013;167:2340-1.
Fontaine JM, et al. Rapid ventricular pacing in a pacemaker patient undergoing magnetic resonance imaging. Pacing Clin Electrophysiol 1998;21:1336-1339.
Forleo GB, et al. Safety and efficacy of a new magnetic resonance imaging-compatible pacing system: Early results of a prospective comparison with conventional dual-chamber implant outcomes. Heart Rhythm 2010;7:750-4
Friedman HL, et al. Magnetic resonance imaging in patients with recently implanted pacemakers. Pacing Clin Electrophysiol 2013;36:1090-5.
Garcia-Bolao I, et al. Magnetic resonance imaging in a patient with a dual chamber pacemaker. Acta Cardiol 1998;53:33-35.
Gimbel JR. Unexpected asystole during 3T magnetic resonance imaging of a pacemaker dependent patient with a ‘modern’ pacemaker. Europace 2009;11:1241-2.
Gimbel JR. Unexpected pacing inhibition upon exposure to the 3T static magnetic field prior to imaging acquisition: What is the mechanism? Heart Rhythm 2011;8:944-5.
Gimbel JR. Letter to the Editor. Pacing Clin Electrophysiol 2003;26:1.
Gimbel JR. Magnetic resonance imaging of implantable cardiac rhythm devices at 3-Tesla. Pacing Clin Electrophysiol 2008;31:795-801.
Gimbel JR. The safety of MRI scanning of pacemakers and ICDs: What are the critical elements of safe scanning? Ask me again at 10,000. Europace 2010;12:915-7.
Gimbel JR, et al. Safe performance of magnetic resonance imaging on five patients with permanent cardiac pacemakers. Pacing Clin Electrophysiol 1996;19:913-919.
Gimbel JR, et al. Outcome of magnetic resonance imaging (MRI) in selected patients with implantable cardioverter defibrillators (ICDs). Pacing Clin Electrophysiol 2005;28:270-3.
Gimbel JR, et al. Randomized trial of pacemaker and lead system for safe scanning at 1.5 Tesla. Heart Rhythm 2013;10:685-91.
Gimbel JR, et al. MRI conditional devices, safety, and access; Choose wisely and when you come to the fork in the road, take Ii. Pacing Clin Electrophysiol 2015 Epub ahead of print.
Gold MR, et al. Preclinical evaluation of implantable cardioverter-defibrillator developed for magnetic resonance imaging use. Heart Rhythm 2015;12:631-8.
Goldsher D, et al. Magnetic resonance imaging for patients with permanent pacemakers: Initial clinical experience. Isr Med Assoc J 2006;8:91-94.
Goldsher D, et al. Successful cervical MR scan in a patient several hours after pacemaker implantation. Pacing Clin Electrophysiol 2009;32:1355-6.
Halshtok O, et al. Pacemakers and magnetic resonance imaging: No longer an absolute contraindication when scanned correctly. Isr Med Assoc J. 2010;12:391-5.
Harden SP. MRI conditional pacemakers: The start of a new era. Br J Radiol 2011;84:773-4.
Hayes DL, et al. Effect of 1.5 Tesla nuclear magnetic resonance imaging scanner on implanted permanent pacemakers. J Am Coll Cardiol 1987;10:782-786.
Heatlie G, Pennell DJ. Cardiovascular MR at 0.5-T in five patients with permanent pacemakers. J Cardiovasc Magn Reson 2007;9:15-9.
Holmes DJ, et al. The effects of magnetic resonance imaging on implantable pulse generators. Pacing Clin Electrophysiol 1986;9:360-370.
Higgins JV, et al. Safety and outcomes of magnetic resonance imaging in patients with abandoned pacemaker and defibrillator leads. Pacing Clin Electrophysiol 2014;37:1284-90.
Inbar S, et al. Case report: Nuclear magnetic resonance imaging in a patient with a pacemaker. Am J Med Sci 1993;3:174-175.
Irnich W, et al. Do we need pacemakers resistant to magnetic resonance imaging? Europace 2005;7:353-65.
International Commission on Non-Ionizing Radiation Protection (ICNIRP) statement, medical magnetic resonance procedures: Protection of patients. Health Physics 2004;87:197-216.
Ipek EG, Nazarian S. Safety of implanted cardiac devices in an MRI environment. Curr Cardiol Rep 2015;17:605.
Ishibashi K, et al. Clinical utility of a magnetic resonance-conditional pacemaker in a patient with cardiac sarcoidosis. Intern Med 2013;52:1341-5.
Jung W, et al. Safe magnetic resonance image scanning of the pacemaker patient: Current technologies and future directions. Europace 2012;14:631-7.
Jung W, et al. MRI and implantable cardiac electronic devices. Curr Opin Cardiol 2015;30:65-73.
Junttila MJ, et al. Safety of serial MRI in patients with implantable cardioverter defibrillators. Heart. 2011;97:1852-6.
Kaasalainen T, et al. MRI with cardiac pacing devices - safety in clinical practice. Eur J Radiol 2014;83:1387-95.
Kaasalainen T, et al. MRI with cardiac pacing devices - safety in clinical practice. Eur J Radiol 2014;83:1387-95.
Kanal E, Barkovich AJ, et al. ACR guidance document for safe MR practices: 2007. Am J Roentgenol 2007;188:1447-1474.
Khan JN, et al. MRI-safe pacemakers and reduction of cardiac MRI artefacts with right-sided implantation. Eur Heart J Cardiovasc Imaging 2013;14:830.
Langman DA, Finn JP, Ennis D. Abandoned pacemaker leads are a potential risk for patients undergoing MRI. Pacing Clin Electrophysiol 2011;34:1051-3.
Langman DA, et al. Pacemaker lead tip heating in abandoned and pacemaker-attached leads at 1.5 Tesla MRI. J Magn Reson Imag 2011;33:426-31.
Langman DA, et al. The dependence of radiofrequency induced pacemaker lead tip heating on the electrical conductivity of the medium at the lead tip. Magn Reson Med 2012;68:606-13.
Lauck G, et al. Effects of nuclear magnetic resonance imaging on cardiac pacemakers. Pacing Clin Electrophysiol 1995;18:1549-55.
Levine PA. Industry Viewpoint: St. Jude Medical: Pacemakers, ICDs and MRI. Pacing and Clinical Electrophysiology 2005;28:266.
Levine GN, Gomes AS, Arai AE, Bluemke DA, Flamm SD, Kanal E, Manning WJ, Martin ET, Smith JM, Wilke N, Shellock FG. Safety of magnetic resonance imaging in patients with cardiovascular devices: An American Heart Association scientific statement from the Committee on Diagnostic and Interventional Cardiac Catheterization. Circulation 2007;116:2878-2891.
Lobodzinski SS. Recent innovations in the development of magnetic resonance imaging conditional pacemakers and implantable cardioverter-defibrillators. Cardiol J 2012;19:98-104.
Loewy J, Loewy A, Kendall EJ. Reconsideration of pacemakers and MR imaging. Radiographics 2004;24:1257-1268.
Luechinger RC. Safety Aspects of Cardiac Pacemakers in Magnetic Resonance Imaging. Swiss Institute of Technology, Zurich, Dissertation, 2002.
Luechinger RC, et al. Pacemaker reed-switch behavior in 0.5, 1.5, and 3.0 Tesla magnetic resonance units: Are reed switches always closed in strong magnetic fields? Pacing Clin Electrophysiol 2002;25:1419-1423.
Luechinger RC, et al. Force and torque effects of a 1.5 Tesla MRI scanner on cardiac pacemakers and ICDs. Pacing Clin Electrophysiol 2001;24:199-205.
Luechinger RC, et al. In vivo heating of pacemaker leads during magnetic resonance imaging. Eur Heart J 2005;26:376-83.
Lowe MD, et al. Safe use of MRI in people with cardiac implantable electronic devices. Heart 2015 Epub ahead of print
Macfie CC, et al. Planned magnetic resonance imaging for a patient with a permanent pacemaker in situ with suspected spontaneous intracranial hypotension. Br J Anaesth 2013;111:1033-4.
Maglia G, et al. Assessing access to MRI of patients with magnetic resonance-conditional pacemaker and implantable cardioverter defibrillator systems: The Really ProMRI study design. J Cardiovasc Med 2015;16:715-20.
Martin TE, Coman JA, Shellock FG, Pulling C, Fair R, Jenkins K. Magnetic resonance imaging and cardiac pacemaker safety at 1.5-Tesla. Journal of the American College of Cardiology 2004;43:1315-1324.
Mattei E, et al. Role of the lead structure in MRI-induced heating: In vitro measurements on 30 commercial pacemaker/defibrillator leads. Magn Reson Med 2012;67:925-35.
Mattei E, et al. Impact of capped and uncapped abandoned leads on the heating of an MR conditional pacemaker implant. Magn Reson Med 2015;73:390-400.
Mollerus M, et al. Ectopy in patients with permanent pacemakers and implantable cardioverter-defibrillators undergoing an MRI scan. Pacing Clin Electrophysiol 2009;32:772-8.
Mollerus M, et al. Magnetic resonance imaging of pacemakers and implantable cardioverter-defibrillators without specific absorption rate restrictions. Europace 2010;12:947-51.
Muehling OM, et al. Immediate and 12 months follow up of function and lead integrity after cranial MRI in 356 patients with conventional cardiac pacemakers. J Cardiovasc Magn Reson 2014;16:39.
Naehle CP, et al. Evaluation of cumulative effects of MR imaging on pacemaker systems at 1.5 Tesla. Pacing Clin Electrophysiol 2009;32:1526-35.
Naehle CP, et al. Safety of brain 3-T MR imaging with transmit-receive head coil in patients with cardiac pacemakers: Pilot prospective study with 51 examinations. Radiology 2008;249:991-1001.
Naehle CP, et al. Magnetic resonance imaging at 1.5-T in patients with implantable cardioverter-defibrillators. J Am Coll Cardiol 2009;54:549-55.
Naehle CP, et al. Safety, feasibility, and diagnostic value of cardiac magnetic resonance imaging in patients with cardiac pacemakers and implantable cardioverters/defibrillators at 1.5 T. Am Heart J 2011;161:1096-105.
Nazarian S, Halperin HR. How to perform magnetic resonance imaging on patients with implantable cardiac arrhythmia devices. Heart Rhythm 2009;6:138-43.
Nazarian S, et al. Clinical utility and safety of a protocol for noncardiac and cardiac magnetic resonance imaging of patients with permanent pacemakers and implantable-cardioverter defibrillators at 1.5 Tesla. Circulation 2006;114:1277-1284.
Nazarian S, et al. A prospective evaluation of a protocol for magnetic resonance imaging of patients with implanted cardiac devices. Ann Intern Med 2011;155:415-424.
Nordbeck P, et al. Measuring RF-induced currents inside implants: Impact of device configuration on MRI safety of cardiac pacemaker leads. Magn Reson Med 2009;61:570-8.
Nordbeck P, et al. Impact of imaging landmark on the risk of MRI-related heating near implanted medical devices like cardiac pacemaker leads. Magn Reson Med 2011;65:44-50.
Nordbeck P, et al. Reducing RF-related heating of cardiac pacemaker leads in MRI: Implementation and experimental verification of practical design changes. Magn Reson Med 2012;68:1963-72.
Nordbeck P, et al. Magnetic resonance imaging safety in pacemaker and implantable cardioverter defibrillator patients: How far have we come? Eur Heart J 2015;36:1505-11.
Quarta G, et al. Cardiovascular magnetic resonance in cardiac sarcoidosis with MR conditional pacemaker in situ. J Cardiovasc Magn Reson 2011;13:26.
Pavlicek W, Geisinger M, et al. The effects of nuclear magnetic resonance on patients with cardiac pacemakers. Radiology 1983;147:149-153.
Pfeil A, et al. Compatibility of temporary pacemaker myocardial pacing leads with magnetic resonance imaging: An ex vivo tissue study. Int J Cardiovasc Imaging 2012;28:317-326.
Pulver AF, et al. Safety and imaging quality of MRI in pediatric and adult congenital heart disease patients with pacemakers. Pacing Clin Electrophysiol 2009;32:450-6.
Quarta G, et al. Cardiovascular magnetic resonance in cardiac sarcoidosis with MR conditional pacemaker in situ. J Cardiovasc Magn Reson 2011;13:26.
Raj V, et al. MRI and cardiac pacing devices - beware the rules are changing. Br J Radiol 2011;84:857-9.
Roguin A, et al. Magnetic resonance imaging in individuals with cardiovascular implantable electronic devices. Europace 2008;10:336-46.
Roguin A, et al. Cardiac magnetic resonance imaging in a patient with implantable cardioverter-defibrillator. Pacing Clin Electrophysiol 2005;28:336-8.
Roguin A, et al. Modern pacemaker and implantable cardioverter-defibrillator systems can be MRI-safe. Circulation 2004;110:475-482.
Russo RJ. Determining the risks of clinically indicated nonthoracic magnetic resonance imaging at 1.5 T for patients with pacemakers and implantable cardioverter-defibrillators: rationale and design of the MagnaSafe Registry. Am Heart J 2013;165:266-72.
Santini L, et al. Evaluating MRI-compatible pacemakers: Patient data now paves the way to widespread clinical application? Pacing Clin Electrophysiol 2013;36:270-8.
Sasaki T, et al. Quantitative assessment of artifacts on cardiac magnetic resonance imaging of patients with pacemakers and implantable cardioverter defibrillators. Circ Cardiovasc Imaging 2011;4:662-70.
Schmiedel A, et al. Magnetic resonance imaging of the brain in patients with cardiac pacemakers. In-vitro- and in-vivo-evaluation at 1.5 Tesla. Rofo 2005;177:731-44.
Shellock FG. Guest Editorial. Comments on MRI heating tests of critical implants. J Magn Reson Imag 2007;26:1182-1185.
Shellock FG. Excessive temperature increases in pacemaker leads at 3-T MR imaging with a transmit-received head coil. Radiology 2009;251:948-949.
Shellock FG. Cardiac pacemakers: Growing evidence for MRI safety. Signals, No. 48, Issue 1, pp. 14-15, 2004.
Shellock FG. Magnetic Resonance Procedures: Health Effects, Safety, and Patient Management CRC Press, LLC, Boca Raton, FL, 2001.
Shellock FG, Crues JV. MR procedures: Biologic effects, safety, and patient care. Radiology 2004;232:635-652.
Shellock FG, Fieno DS, et al. Cardiac pacemaker: In vitro assessment of MR safety at 1.5-Tesla. American Heart Journal 2006;151:436-443.
Shellock FG, Fischer L, Fieno DS. Cardiac pacemakers and implantable cardioverter defibrillators: In vitro evaluation of MRI safety at 1.5-Tesla. Journal of Cardiovascular Magnetic Resonance 2007;9:21-31.
Shellock FG, et al. Cardiac pacemakers and implantable cardiac defibrillators are unaffected by operation of an extremity MR system. Am J Roentgenol 1999;72:165-170.
Shellock FG, Tkach JA, Ruggieri PM, Masaryk TJ. Cardiac pacemakers, ICDs, and loop recorder: Evaluation of translational attraction using conventional (“long-bore”) and “short-bore” 1.5- and 3.0-Tesla MR systems. Journal of Cardiovascular Magnetic Resonance 2003;5:387-397.
Shellock FG, Valencerina S, Fischer L. MRI-related heating of pacemaker at 1.5- and 3-Tesla: Evaluation with and without pulse generator attached to leads. Circulation 112;Supplement II:561, 2005.
Shinbane J, Colletti P, Shellock FG. MR in patients with pacemakers and ICDs: Defining the issues. Journal of Cardiovascular Magnetic Resonance 2007;9:5-13.
Shinbane J, Colletti P, Shellock FG. MR imaging in patients with pacemakers and other devices: Engineering the future. Journal of the American College of Cardiology Imaging 2012;5:332-3.
Shinbane J, Colletti P, Shellock FG. Magnetic resonance imaging in patients with cardiac pacemakers: Era of “MR conditional” designs. Journal of Cardiovascular Magnetic Resonance 2011;197:W457-9.
Sommer T, Vahlhaus C, et al. MR imaging and cardiac pacemakers: In vitro evaluation and in vivo studies in 51 patients at 0.5 T. Radiology 2000;215:869-879.
Sommer T, et al. Strategy for safe performance of extrathoracic magnetic resonance imaging at 1.5 Tesla in the presence of cardiac pacemakers in non-pacemaker-dependent patients: A prospective study with 115 examinations. Circulation 2006;114:1285-92.
Sommer T, et al. German Roentgen Society Statement on MR Imaging of Patients with Cardiac Pacemakers. Rofo 2015;187:777-787.
Strach K, et al. Low-field magnetic resonance imaging: Increased safety for pacemaker patients? Europace 2010;12:952-60.
Sutton R, Kanal E, Wilkoff BL, et al. Safety of magnetic resonance imaging of patients with a new Medtronic EnRhythm MRI SureScan pacing system: Clinical study design. Trials 2008;9:68.
Tandri H, et al. Determinants of gradient field-induced current in a pacemaker lead system in a magnetic resonance imaging environment. Heart Rhythm 2008;5:462-8.
Ubee SS, et al. Implications of pacemakers and implantable cardioverter defibrillators in urological practice. J Urol 2011;186:1198-205.
Vahlhaus C, et al. interference with cardiac pacemakers by magnetic resonance imaging: Are there irreversible changes at 0.5 Tesla? Pacing Clin Electrophysiol 2001;24(Pt. I):489-95.
van der Graaf AW, et al. MRI and cardiac implantable electronic devices: Current status and required safety conditions. Neth Heart J 2014;22:269-76.
Verma A, et al. Canadian Heart Rhythm Society and Canadian Association of Radiologists consensus statement on magnetic resonance imaging with cardiac implantable electronic devices. Can J Cardiol 2014;30:1131-41.
Weishaupt D, et al. Pacemakers and magnetic resonance imaging: Current status and survey in Switzerland. Swiss Med Wkly 2011;141:w13147.
Wilkoff BL, et al. Magnetic resonance imaging in patients with a pacemaker system designed for the magnetic resonance environment. Heart Rhythm 2011;8:65-73.
Wilkoff BL, et al. EnRhythm MRI SureScan Pacing System Study Investigators. Magnetic resonance imaging in patients with a pacemaker system designed for the magnetic resonance environment. Heart Rhythm 2011;8:65-73.
Wilkoff BL, et al. Safe MRI scanning of patients with cardiac rhythm devices: A role for computer modeling. Heart Rhythm 2013;10:1815-21.
Wollmann C, et al. Safe performance of magnetic resonance imaging on a patient with an ICD. Pacing Clin Electrophysiol 2005;28:339-42.
Wollmann CG, et al. Clinical routine implantation of a dual chamber pacemaker system designed for safe use with MRI: A single center, retrospective study on lead performance of Medtronic lead 5086MRI in comparison to Medtronic leads 4592-53 and 4092-58. Herzschrittmacherther Elektrophysiol 2011;22:233-6, 239-42.
Wollmann CG, et al. Monocenter feasibility study of the MRI compatibility of the Evia pacemaker in combination with Safio S pacemaker lead. J Cardiovasc Magn Reson 2012;14:67.
Wollman CG, et al. Safe performance of magnetic resonance of the heart in patients with magnetic resonance conditional pacemaker systems: the safety issue of the ESTIMATE study. J Cardiovasc Magn Reson 2014;16:30.
Zikria JF, et al. MRI of patients with cardiac pacemakers: A review of the medical literature. Am J Roentgenol 2011;196:390-401.
Zimmermann BH, Faul DD. Artifacts and hazards in NMR imaging due to metal implants and cardiac pacemakers. Diagn Imaging Clin Med 1984;53:53-56.